Some USB-to-serial converters may cause data loss, if they do not keep up with the data flow. This is very rare. When it occurs, the BrainMaster detects the data loss, and will continue to process when good data arrive. It is possible to see if this has occurred by using the “Data / Login Data” control panel to see if any “resyncs” have occurred. In the older software, this could produce a possible “sawtooth” appearance of the waveforms, that is readily visible. If you see this, discontinue training. To prevent it, make sure you have the current version of your software.
Powered by BetterDocs
Your email address will not be published. Required fields are marked *
By using this form you agree with the storage and handling of your data by this website. *